1. Electricity Connection
You need to make sure the power supply you are going to connecting the tester with meets requirements listed on the information label of the tester.
And also, if the tester comes with a power cable, connect correct leads to the power system.
2. Battery Connection
Each channel of Neware battery tester has at least four sensing terminals – I+/V+ and I-/V-.
- The voltage sensing should not be connected on the current leads.
- The current sensing should be connected tightly to improve current passing capability.
Custom battery fixture available from us.
3.1 Software - LAN-based System
Neware battery tester functions a LAN device, using TCP/IP to transfer data. Additionally there is SERVER IP to be configured on the tester, identifying only the hosting computer having an IP address of SERVER IP can control this tester.
3.2 Software - Software Installation
Software Installation & Configuration
You may refer to below link for software package.
General BTS Server/Client
After the installation, you will have BTS Server and BTS Client on the computer. You may also deploy BTS client and BTS server on separated computers.
Advanced Reporter/MES/LIMS for different scenarios.
Refer to System Feature for more
3.3 Software - Channel Mapping
After you have SERVER IP set, you will see connected testers on the left tree menu from BTS Client. You still need to map the channels before you can operate each one.
- Click on the target number of the tester.
- Choose Reset Map(M) *on the right click menu of the right area from BTS Client.
- You can choose 1*8 or 1*16 for Main channel mapping.
3.4 Software - Start a Test
After channels be mapped, you are now able to start a test on selected channels.
Work steps can also be saved as a step file in XML file format for future load.
3.5 Software - Data View
You can use View Data on right click menu of the selected channel to view data. This will download data from the server to your local file system and launch BTSDA to open.
BTSDA opens NDA data, rendering data in 2 parts – left is the plotted curves and right is detailed records.